COM Research Information: -80 Freezers and Incubator Monitoring (Dickson/Oceasoft)
Dickson (formerly Oceasoft) is the current temperature freezer monitoring company the College of Medicine is working with along with our COM departments.
For those of you that oversee research activities, the COM Dean's Office is in the process of installing Oceasoft Gateways to help with remote monitoring of -80 freezers and incubators as part of the MSB capital project. The gateways will be strategically deployed to provide optimal/equitable coverage in the COMRB, MSB, and MBRB. Departmental research programs will be able to leverage the infrastructure by purchasing the Oceasoft equipment modules at a reasonable price and then monitor the equipment via a cloud based platform/app. If the initial program proves productive, we will exam a broader deployment of the infrastructure to other facilities.
1. In order to leverage these new gateway receivers, older technology would need to be upgraded.
2. Electrical power and a data jack (under COM VLAN) is required.
3. COM will be the admin for the Oceasoft cloud portal and will create the depts and accounts for users.
4. Any equipment outside the gateway is the responsibility of the depts.
5. Firmware, software, cloud, hardware, software is the responsibility of Dickson to solve.
6. COM is only responsible for gateway hardware, network connection and item 3.
General layout: sensors are put inside freezers which connect to data loggers that are usually located on the freezer and data logger sends freezer data via wireless to gateways which are usually mounted on a wall (or sitting on a shelf) and connected to power and a data jack. Gateways send data to the cloud and accessible via Dickson portal.
Dickson/Oceasoft contact info: Adrien Sedillot <email@example.com>
https://www.oceasoft.com/files/library/oceaview/oceasoft_lorawan_receiver_user_guide_en.pdf (we have FW v. 5.1.x).
COM contact info: firstname.lastname@example.org
LoRaWAN gateway locations and details.
BLDG, room#, datajack#, mac address, assigned IP, state tag:
MSB E511: G-108, 00:08:00:4A:F7:13. 10.108.1.42. ptag: P20228166.
COMRB 8040: A021, 00:08:00:4B:04:54. 10.109.93.47 ptag: P20228967.
COMRB 4060: E092, 00:08:00:4B:04:3C. ptag: P20228963.
COMRB 5140: D-2359 data 1 active with network switch use.00:08:00:4A:B9:60. ptag: P20223498.
COMRB 2040: G047, 00:08:00:4B:04:39. ptag: P20228962.
MBRB 3205: W-D087, 00:08:00:4B:04:5F. 10.108.1.28 assigned to former gateway. ptag: P20228966. fw updated
MBRB 3115: W-C122, 00:08:00:4B:04:5C. 10.108.1.27 . ptag: P20228960.fw updated.
MBRB 2205 00:08:00:4B:04:3D. ptag: P20228959. fw updated.
MBRB 2115 00:08:00:4B:04:60. ptag: P20228961. fw updated.
SPARE: 00:08:00:4C:58:6B. ptag: P20228958. fw updated. (replaced 00:08:00:4B:04:55)
SPARE: 00:08:00:4B:04:3B. ptag: P20228965. fw updated.
SPARE: 00:08:00:4B:04:3A. ptag: P20228964. fw updated.
Oceasoft equipment and services exist on campus outside of what we have here but we are not responsible for it.